BORK operational status:
The station operator visited the station and installed a new Aux box, term box, and ADSL DC converter.
The station operator visited the station and installed MCG surge protection.
The BORK10 Q330 was frequently rebooting since late 2019. On 11/24/2020, the Q330 cabling was swapped to BORK00 Q330 and then that Q330 was reconfigured as BORK10. No Q330 reboots since swap.
The station operator visited the station and replaced the station computers on day 2020,295-09:00.
The station operator moved the Episensor to the BORK10 Q330 digitizer and powered off the BORK00 digitizer until power issues are resolved and the faulty STS-6 is replaced.
The station is back online after the station operator reconfigured the connections on the Advantech switch and restored power to 4 of the 12VDC ports on the remote power supply.
The station is down as of day 2020,206-09:24 (July24) due to failure of the remote power supply. IDA engineers are preparing a shipment of power system equipment to restore power at the station.
Data quality for all channels from the STS-6 is poor as of day 2020,162-04:27. IDA engineers and the sensor manufacturer are investigating this issue.
New IDA station BORK in Borovoye, Kazakhstan installed with a Quanterra Q330 data acquisition system. Sensors installed include STS-6 and STS-5 borehole seismometers, and an Episensor strong motion detector.